Sign up or log in Sign up using Google. If there are no data points available, the value 9. The example programs are provided solely for the purpose of illustrating a subset of the SCPI Standard Commands for Programmable Instruments commands used with the A. Note that the data is transfered as ASCII, and thus it is not recommended to transfer a large number of data points using this method. Measurements will begin when specified triggering conditions are met, following the receipt of the INIT command.
|Date Added:||27 February 2018|
|File Size:||31.83 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
Measurements will begin when specified triggering conditions are met, following the receipt of the INIT command. These same programs are also included in the Instrumnet Basic download.
Agilent A instrument driver – MATLAB & Simulink
The latest version can be downloaded from: Sign up using Facebook. Legal Privacy Terms Trademark Acknowledgements. Downloads an arbitrary waveform to the generator as binary instrumebt. Readings are NOT erased from memory when using fetch.
Permanent clearance sale – Keysight / Agilent 33220A
VBA code isn’t written with Visual Studio. For faster downloads, use BinaryArb.
This is similar to calling init and then immediately following fetch. Trigger source for this example is the bus. This model has been replaced by the Keysight series waveform generators. Example programs for the A may be downloaded below. Learn 333220a control instruments with Visual Basic: Please don’t tag with vba questions that have nothing to do with VBA.
The edge time is incremented.
Agilent — InstrumentKit Library documentation
This instrjment may or may not work with these newer models. They are organized into different zip files, depending upon the programming environment:.
Sets start and stop frequency, as well as sweep time. You can find the full specifications for these instruments on the Keysight website. Symmetry sgilent the amount of time per cycle that the ramp wave is rising unless polarity is inverted.
There is also a separate zip file for those programs that are published in the A User’s Guide.
Have the multimeter perform a specified number of measurements and then transfer them using a binary transfer method. This is probably off-topic for this site. Sign up or log in Sign up using Google.
First data point sent to output buffer is the oldest. Very easy to interface. Data will be cleared from instrument memory after transfer is complete. Creates a linear sweep for a sinusoid.
The instrument has a fixed series output impedance of 50ohms. The example programs are provided solely for the purpose of illustrating a subset of the SCPI Standard Commands for Programmable Instruments commands used with the A.